The Model 106-H is a high concentration ozone monitor designed for measuring and controlling output concentration of a high concentration ozone generator. Ozone is measured in the range 0-20 wt% with a precision of 0.01 wt% or 2% of reading by the method of UV absorbance. The instrument has the option of a water-proof NEMA enclosure.
|CUV-1 Carrying Case||This is an extremely rugged pelican case to carry your UV-106 in.|
|ODS-1 Destruct Unit||This is an inexpensive ozone destruct unit, made for removing high concentrations of ozone efficiently.|
|FM-6-2 Ozone Compatible Flowmeter||Designed for specifically for corrosive ozone gas flow-rates, this flow-meter is a precision instrument embodying the inherent simplicity, versatility and economy of the classical variable area meter.|
- High Range Ozone Measurement
- Compact size, bench-top design
- Flow-through design ideal for in-line ozone measurement
- Benchtop, NEMA & OEM configurations available
|Electrical||11 - 28 VDC nominally 440 mA at 12V, 5.3 watt|
|Dimensions||8.5 x 8.5 x 3.8 inches (21.6 x 21.6 x 9.5 cm)|
|Detection Method||UV Absorption|
|Temperature Range||32 to 122°F (0.0 to 50°C)|
|Accuracy||0.01 wt% or 2 % of reading|
|Resolution||0.01 Percent By Weight|
|Supporting Documents||User Manual & NIST Traceable Calibration Sheet Included|
|Internal Datalogging||Yes, 32,736 Lines|
|Display Units||wt%, mol%|
|Communication Protocol||RS-232, USB|
|Zero Drift||1% of full scale/month, non-cumulative|
|LPM Flow Rate||1 LPM|
|Portability||Yes , with purchase of B3 Battery.|
What is the maximum amount of tubing I can use to transport ozone to the analyzer?
30 feet of tubing would be the maximum length of tubing recommended.
I cannot seem to turn datalogging off on my analyzer. Every time I turn the device on, it resumes datalogging even if I have previously turned it to "off" in the Menu.
This is a bug that occasionally occurs. The following procedure clears up the problem: 1) "Transmit" existing data (this may take some time) 2) "End" the data logging session 3) Log 4 or 5 new data points 4) "End" the data logging session 5) Turn the instrument off When you restart the device, data logging should be off and should stay off - if not, try repeating steps 4 - 6 again. (Please refer to the Operation Manual for your device if you have questions on how to "Transmit", start or end datalogging sessions).
Does this monitor have CE certification?