The Model 106-H is a high concentration ozone monitor designed for measuring and controlling output concentration of a high concentration ozone generator. Ozone is measured in the range 0-20 wt% with a precision of 0.01 wt% or 2% of reading by the method of UV absorbance. The instrument has the option of a water-proof NEMA enclosure.
CUV-1 Carrying Case
This is an extremely rugged pelican case to carry your UV-106 in.
ODS-1 Destruct Unit
This is an inexpensive ozone destruct unit, made for removing high concentrations of ozone efficiently.
FM-6-2 Ozone Compatible Flowmeter
Designed for specifically for corrosive ozone gas flow-rates, this flow-meter is a precision instrument embodying the inherent simplicity, versatility and economy of the classical variable area meter.
High Range Ozone Measurement
Compact size, bench-top design
Flow-through design ideal for in-line ozone measurement
Benchtop, NEMA & OEM configurations available
11 - 28 VDC nominally 440 mA at 12V, 5.3 watt
8.5 x 8.5 x 3.8 inches (21.6 x 21.6 x 9.5 cm)
32 to 122°F (0.0 to 50°C)
0.01 wt% or 2 % of reading
0.01 Percent By Weight
User Manual & NIST Traceable Calibration Sheet Included
What is the maximum amount of tubing I can use to transport ozone to the analyzer?
30 feet of tubing would be the maximum length of tubing recommended.
I cannot seem to turn datalogging off on my analyzer. Every time I turn the device on, it resumes datalogging even if I have previously turned it to "off" in the Menu.
This is a bug that occasionally occurs. The following procedure clears up the problem: 1) "Transmit" existing data (this may take some time) 2) "End" the data logging session 3) Log 4 or 5 new data points 4) "End" the data logging session 5) Turn the instrument off When you restart the device, data logging should be off and should stay off - if not, try repeating steps 4 - 6 again. (Please refer to the Operation Manual for your device if you have questions on how to "Transmit", start or end datalogging sessions).
Does this monitor have CE certification?
Seller warrants that all Products will conform to the Specifications and will be free from defects in material workmanship and programming. Claims for Products not complying with this warranty shall be submitted by Buyer no later than 12 months from delivery date to Buyer. Seller shall promptly repair or replace, at its option and expense, non-complying Products. The shipment of non-complying Products by Buyer to Seller, and the return shipment of repaired or replacement Products by Seller to Buyer shall be at Seller's expense, unless it shall be determined that Buyer returned a Product that was in compliance, in which case such cost shall be borne by Buyer and, if paid by Seller, these expenses shall be reimbursed by Buyer within thirty (30) days of notice by Seller. Buyer shall receive a credit for any non-complying Products returned for repair or replacement. All non-complying Products shall be returned to Seller (at its offices or as otherwise instructed) no later than 30 days as of the date the non-compliance of any such Product has been discovered. Seller is NOT responsible for any/all damages suffered to the Products that is caused by the Buyer. Such as but not limited to damage by wrong electrical current, damage through crush or dropping the product, damage through untrained users of Product, damage through improper operations.
California Regulatory Warning
All of Ozone Solutions ozone generators are intended for industrial use ONLY.
Use in unoccupied spaces only.
Health hazard warning: emits ozone.
This product is not CARB Certified (Ozone Emissions for Air Cleaners).
Product Disclaimer: Does not meet California air cleaner regulation requirements; cannot be shipped to California.